- Buyers Guide
Aerospace & Defense Electronics Supplement
Early Returns: U.S. Export Control Reform Positive
A&D Test & Measurement
Efficient Design and Analysis of Airborne Radomes
Rohde & Schwarz has developed a new method that accurately measures group delay and phase linearity on satellite links. With the new R&S ZVA-K10 software option, users can perform measurements using two network analyzers located at separate sites. In open-area tests, for example, the analyzers can bridge long distances between a transmitter and a receiver. Also, the network analyzers do not need to access the local oscillator when measuring frequency-converting DUTs. The R&S ZVA-K10 option is available for the R&S ZVA and R&S ZVT high-end network analyzers.
Precise measurement of group delay and phase linearity is vital in order to determine satellite communications quality. Today's ever-increasing transmission rates also place more stringent demands on transmission quality. Key parameters for transmission quality are group delay and phase linearity within the useful band. With the test setup from Rohde & Schwarz, users can perform accurate measurements on upconverters and downconverters used in satellite communications, as well as on complete transmission systems or individual components such as transmitters, receivers or mixers.
The conventional approach to measuring group delay and phase linearity is to connect the input and output of a DUT to a network analyzer via RF cables. However, if the input and output are located far apart, cable losses will degrade the measurement's signal-to-noise ratio. In open-area measurements, cables need to bridge distances of several hundred meters. In addition to causing high loss, cables also introduce significant phase errors even if they are just slightly moved.
The new test method from Rohde & Schwarz provides reliable results. In a test setup with the R&S ZVA-K10 software option, the user connects one network analyzer to the transmit port and one to the receive port. The two network analyzers communicate with each other and synchronize the test sequence between each other via a LAN connection.
With regards to the DUT, the R&S ZVA stimulates the DUT with a two-tone signal and measures the phase difference between the two carriers at the input and the output. From the phase difference, the R&S ZVA calculates the group delay and the relative phase of the DUT's transfer function. Any changes in the local oscillator's frequency and phase have an identical effect on both carriers and are eliminated by the difference measurement; they do not, therefore, affect measurement accuracy.
Get access to premium content and e-newsletters by registering on the web site. You can also subscribe to Microwave Journal magazine.